Read online or download PDF • Page 44 / 67 • KROHNE IFC Converter EN User Manual • KROHNE Equipment. Micro Precision provides expert calibration services for KROHNE instruments including KROHNE IFC D -. Select from options for the IFC D which. KROHNE 08/ Subject to change without notice. Applicable to. Software-Versions. ○ IFC _. / D. Display version. No. and.
|Published (Last):||22 January 2010|
|PDF File Size:||12.81 Mb|
|ePub File Size:||13.22 Mb|
|Price:||Free* [*Free Regsitration Required]|
Tell us about it. If installed and operated in accordance with these operating instructions, it will rarely present any problems. Switches, counters, ufc ys and recorders. Underwater and underground installation possible. The selected creepage distances and clearances have been sized in conformity with VDE and IEC for contamination category 2.
Setting ranges The ranges are dependent on the meter size DN and the flow velocity v: Page 11 Standard factory settings Function 1.
Page 3 1 Electrical connection: Don’t see a manual you are looking for?
KROHNE IFC 010 Converter Short EN User Manual
Tell us what’s missing. Page 14 Function 1. Signal cable to be permanently laid.
Supply circuits and output circuits are designed to meet the standards of korhne classes III and II, respectively. Do not cross or loop cables inside the terminal compartment. Page 8 Connection of shields to the IFC 1 DS Signal cable 2 Shield terminal for shielded signal cable 3 Field power cable 4 Shield terminal for shielded field power cable 5 Housing wall, signal converter 6 Cable entries 2 Electrical connection of outputs 2. The flowmeter housings protecting the electronic equipment from dust and moisture must always be kept closed.
KROHNE IFC Converter Short EN User Manual | 19 pages
Page 10 3 Start-up 3. Your manual failed to upload Page 2 Contents 1 1. P Pulse output overranged. Page 15 Function 3.
Passive mode External power source required for operation of the outputs. All operating data have been factory set in accordance with your specifications. Power the unit, and the flowmeter will immediately start process flow measurement. Or point us to the URL where the manual iffc located.
Use separate cable entries see below for power supply, field current cables, signal lines, outputs and inputs.
Connection of shields to the IFC 1 DS Signal cable 2 Shield terminal for shielded signal cable 3 Field power cable 4 Shield terminal for shielded field power cable 5 Housing wall, signal converter 6 Cable entries 2 Electrical connection of outputs 2.
Vo r tex flowmeters. Lev el measuring instruments. These concise instructions do not include the following: Connect shields via stranded drain wires. V ariable area flowmeters. The operator is, howeverobligated to take note of these sections in the.